|March 26, 2016|
Actigraphy is a relatively non-invasive method of monitoring human rest/activity cycles. A small actigraph unit, also called an actimetry sensor, is worn by a patient to measure gross motor activity. Motor activity often under test is that of the wrist, measured by an actigraph in a wrist-watch-like package. The unit continually records the movements it undergoes. The data is later read to a computer where it can be analysed.
Sleep actigraphs are generally watch-shaped and worn on the wrist of the non-dominant arm. They are useful for determining sleep patterns and circadian rhythms. They may be worn for weeks at a time.
Actigraphy is useful for assessing daytime sleepiness in situations where a laboratory sleep latency test is not appropriate. It is used to clinically evaluate insomnia, circadian rhythm sleep disorders, excessive sleepiness and restless legs syndrome. It is also used in assessing the effectiveness of pharmacologic , behavioural, phototherapeutic or chronotherapeutic treatments for such disorders.
Actigraphy has not traditionally been used in routine diagnosis of sleep disorders but is increasingly being employed in sleep clinics to replace full polysomnography. The technique is more extensively used in academic research and is being increasingly employed in new drug clinical trials where sleep quality is seen as a good indicator of quality of life.
Activity actigraphs are worn and used similar to a pedometer: around the waist, near the hip. They are useful for determining the amount of activity and possibly the number of calories burned by the wearer. They are worn for a number of days.
Movement actigraphs are generally larger and worn on the shoulder of the dominant arm. They contain a 3D actigraph as opposed to a single dimension one, and have a high sample rate and a large memory. They are used for only a few hours, and can be used to determine problems with gait and other physical impairments.
The unit itself is an electronic device which generally consists of:
Actigraphs have a number of different ways of accumulating the values from the accelerometer in memory. ZCM (zero crossing mode) counts the number of times the accelerometer waveform crosses 0 for each time period. PIM (proportional integral mode) measures the area under the curve, and adds that size for each time period. TAT (time above threshold) uses a certain threshold, and measures the length of time that the wave is above a certain threshold. Literature shows that PIM provides most accurate measurements for both sleep and activity, though the difference with ZCM is marginal.
Actigraph units vary widely in size and features and can be expanded to include additional measurements. However, there are a number of limiting factors:
For some uses, the following are examples of additional features:
Some modern alarm clocks use an actigraph to identify periods of lighter sleep , when the sleeper should wake more easily. An application for Apple's iPhone, Sleep Cycle alarm clock, generates graphs of body movement during sleep and has a similar alarm clock function.
GNU Free Documentation License. It uses material from the Wikipedia article "Actigraphy".
All informatin on the site is © www.diseases-diagnosis.com 2002-2011. Last revised: January 2, 2011|
Are you interested in our site or/and want to use our information? please read how to contact us and our copyrights.
To let us provide you with high quality information, you can help us by making a more or less donation: